Model Name : W/Pc2_M
Application fields: Vegetable greenhouse temperature and humidity detection、storage temperature and humidity detection、industrial environment temperature and humidity detection, etc
1.Device realizes real-time collecting temperature and humidity data.
2.Software realizes real-time reading environment temperature and humidity, and saves the data.
3.Software supports the Windows system
4.Standard POE power supply mode, compatible with 802.3 af
New Functions !!
(1) 1W340 products newly added Android APP, support all Android phones and tablets. Just download this software1W340_Android_V1.0.zip,
1W340 product Newly add remote access function , welcome to download the newest software (1W340 V4.0 zip) .
Operating steps: 1. Search Device >> 2. Register Account >>3. Login user >> Set device config >>4. Search network device
The default LAN basic configuration
1.Working voltage: DC 5V
3.Equipment power: < 0.6 W
(1)IP address: 192.168.1.188 (the user can modify)
(2)The subnet mask: 255.255.255.0 (the user can modify)
(3). The gateway: 192.168.1.1 (the user can modify)
(4). Connection types: TCP (default shall not modify)
(Device supports secondary development, the details please consult the salesman)
Device using aviation plug connecting internal circuit board, probe and network port. Please tighten the aviation plug, and then connect the switch that support POE with the cable. Device can work( Note: You must insert the plug before power-on).
2.1 Panel function
The front of the device has 8 lights to indicate the working status and a button to restore factory settings, function as follows:
POWER light on : power supply normal
LINK light on: network connecting normal
ACK light flash: network port is sending and receiving data
RELAY light: Indicator relay status，when the relay closes the light is on or the light is off
CH1~CH4 light: If the device reads normally the light will flash.
2.2 Direct connection
Connect device to computer with one cable, generally called direct connection or point-to-point connection. It makes it easy to visit device, wire and opera.
Note: The device supports POE power supply mode only.
2.2.1 Network Settings
Factory default IP (192.168.1.188) subnet mask (255.255.255.0) gateway (192.168.1.1).Computer and device must in the same network segment .
Steps as follows:
Step one: "Network Place" -> 'Check the local connection" ->"Local Connection", right click the "Local Connection"->"Properties"
Step Two: "This connection uses the following items" ->"Internet Protocol Version (TCP/IP)"->"Properties"
Step three: Select "Use the following IP address", fill in the subnet mask, default gateway same to the device. Set the IP as 192.168.1.xxx, the final data (xxx) of device IP and the computer IP can not be same (note : xxx don't set 0 or 255)，Set up and then click "ok", save.
2.3 Networking connection
The device and the computer have only one network port, if the user wants to visit several devices, or use several computers to visit one or more devices, it requires the user to form a network, the user needs to choose the switch supports POE. There must be more switch ports than the devices.
If the computer and the device is not in the same network segment, users need to modify the IP, subnet mask, gateway to fit with current network, please refer to the device software setting
2.4 Restore factory settings
If you forget the parameters or incorrectly set to make it not work, you can press the Preset button on device for 3~5 seconds. It is finished when the LINK and ACK light go out.
3. Data detection
The resolution and precision of the data will be different by different probes. Please refer to the table below
4.1. Search probe
Open software will appear the following window, fill in equipment starting the IP address and ending IP address, then , software will search the device from starting to end the IP address .(including all the 1 W340 equipment )
4.2. Real-time temperature
Click on "real-time data” and select IP address what you want to check the equipment. As follow:
Choose what you want to see the equipment IP address and read the temperature (As follows). log default save directory: my document PCsensor 1 W340 V1.3 Data
Choose what you want to see the equipment IP address
Click left mouse button under marquee can zoom curve and narrow curve; Click and hold the right mouse button, can move the curve.
Click on “Setting” and select “General Setting “. As follows:
Click"Setting" and choose "Device Setting" , choose one IP address, as follow:
Click on the 18 b20, will enter DS18b20 temperature probe to set temperature calibration value.
4.6. MSN remote access to the temperature
MSN users log in, your friends can send related instructions and to obtain the relevant temperature data. When your friends send prompted number 1 to you, he will automatically get the temperature data. Including: real-time temperature, maximum, minimum and average.
4.7. Email function
Fill-mail message, and save. (Send E-mail request: support for POP3/SMTP services or other ports.) Receiving mail can fill two.
Can realize the following functions:
1, open warning function, when the alarm is automatically triggered to send a message to the specified mailbox;
2, you can set the time, from time to time, it will send a message to the specified mailbox (recommended);
3, open continuous send, You can send mail to open a row in the specified mailbox (not recommended).
Waterproof network Specification(English)-2014
beta version v3
Copyright© 2014 by SHENZHEN RDING TECH CO., LTD.
Note: If you want to download the latest software, please click here： http://software.pcsensor.com/pc_en.html
Product Code: W/Pc2_M